The OrangeLink is a compact, lightweight, RileyLink compatible device designed using an all new nRF52810 Bluetooth 5.2 System on Chip (SoC). The OrangeLink includes both 433MHz and 915-866MHz antenna’s, and uses 2 AAA batteries for 2-3 weeks of runtime without having to charge the device. The OrangeLink also includes a vibration feature that notifies the user when the battery is low or if the OrangeLink is disconnected from a phone app. The design matches the same dimensional specifications as the Apple AirPods™ and fits many of the AirPods™ Protective cases and accessories. Please Join the OrangeLink Facebook Group for Discussions
We are proud to announce that we are working with Tidepool.org. We are donating a portion of proceeds from the OrangeLink directly to Tidepool.org to help them bring Tidepool Loop to the T1d community.
The OrangeLink is a completely new RileyLink compatible design by Vic Wu (Birdfly). The OrangeLink uses 2 AAA batteries for up to 4 weeks runtime*. The OrangeLink has totally new Bluetooth and RF processors that allow users to update the firmware from their mobile phones as new features and firmware updates are released. Both Medtronic(915-866MHz) and Omnipod(433MHz) antennas are on board, and a vibration notification feature to alert of App Disconnects and Battery Low Level.
nRF52 – N52810 Nordic Semiconductor SoC ARM Processor
64 MHz Cortex-M4
192 KB Flash, 24 KB RAM
2 AAA Batteries (~11-14 days Lithium, ~7-10 days Alkaline, ~7-10 days Rechargeable Lithium)
Vibration Notifications for Loop App Disconnect, and Battery Low Status
Both 915-866MHz (Medtronic) and 433MHz (Omnipod) antennas on board
Apple AirPod™ Case Form Factor (can use AirPod™ Case protectors/holders)
Updatable Firmware over Bluetooth
% Battery Status, and Battery Low Warning (to be added to Loop/AndroidAPS in future App updates)
Included in your order
USPS First Class USA Shipping
USPS First Class International Shipping Included (up to 30 days due to COVID19)
(You may need to pay your Gov’t Customs Import Tax when you receive the package)
DHL Express International – available at Checkout for 3-5 day international shipping.
运往中国或香港的货物需要使用转运服务xianfengex.com / myus.com
Shipments to China or Hong Kong require the use of a Trans-Shipment Service
We will make every effort to ship your RileyLink to you as quickly as possible.
1-3 Days – You will receive an email with a Shipment Tracking Number
1-2 Days Later – You will start to see the shipment moving in USPS System
2 Days Later (USA Shipments)- should be delivered (7 Days Total)
10-15 Days Later (International Shipments) should be delivered. Customs seems to randomly hold packages so you may need to call your country’s postal customs office. (20 Days Total)
4.) To RESET the OrangeLink. Open the battery lid and then close it again.
LED and Haptic Vibrator Feedback Indications
When OrangeLink is first powered on, LED light will blink Red and Yellow with a strong vibration.
When OrangeLink is powered on but not connected to a phone the LED light will blink Red every 10 seconds
When OrangeLink is first connected to a mobile phone, a SOFT vibration is triggered.
When the OrangeLink is connected to a mobile phone, LED light will blink Yellow every 10 seconds
When OrangeLink is disconnected from a mobile phone a STRONG vibration is triggered.
When battery is low, LED light will be constant RED, and with SOFT vibration every 2 mins.
When the OrangeLink has a “super low” battery, the voltage is too low to trigger constant RED and SOFT vibration, you will see the OrangeLink repeatedly (connect then disconnect) from the phone (See below for Super Low Indication)
MySentry Disable for Medtronic 754/723/554/523: Medtronic x54 and x23 pumps have a feature called MySentry which initiates broadcasts from the Pump at predetermined intervals. This requires the OrangeLink or any RileyLink Compatible device to “listen” continuously for these messages. This “listening” for messages prevents the OrangeLink from powering-off the Radio chips on the board to conserve power. With the power to the radio chips on all the time the OrangeLink will only last a few days on a pair of batteries. It is highly recommended if you’re using a Medtronic 754/723/554/523 pump to disable MySentry in your Apps Code.
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.